Are you looking for the Venturi Meter? So today we will study the Definition, Parts, Working Principle, Diagram, Experiment, Derivation, Formula, Application, Advantages, Disadvantages, PDF. Here you will get the articles of Mechanical Engineering in brief with some key points and you will get to know an enormous amount of knowledge from It. So If you find this articles helpful, please let us know in the comment box, either if any correction required too, also let us know in the comment box.
Venturi Meter is a device or component which is used for measuring the rate of flow of liquid through pipes.
The rate of Fluid = Discharge.
Venture Meter is first developed by G.B. Venturi in 1797.
It works on the basic principle of Bernoulli’s Equation.
Bernoulli’s Equation: In Simple, this is defined as the sum of Pressure energy, Kinetic energy per unit volume and Potential energy per unit volume equal in the same direction.
Venturi Meter Definition:
Venturi Meter is a device in which pressure energy is converted into kinetic energy and It used for measuring the flow rate (Discharge) of the system.
Venturi Meter Parts:
It consists of 3 parts:
- Converging cone or Diameter (the area is Decreasing)
- Throat Diameter (the area is constant)
- Diverging cone (the area is increasing)
let’s consider a pipe in which there is venturi meter is fixed. In the pipe fluid is flowing so first it enters into a converging cone then Thorat and then Diverging Cone.
When water flowing through this cone the area is decreasing, therefore, the speed of flowing water increases and pressure decreases.
When water flowing through this cone the area remains constant therefore the speed of flowing water and pressure remains constant.
When water flowing through this cone the area is increasing, therefore, the speed of flowing water decreases and pressure decreases.
Venturi Meter Working Principle / Diagram:
Venturi Meter works on the basic principle of Bernoulli’s Equation.
Venturi Meter Derivation / Formula:
The derivation has been solved on paper because their several rotations cannot be typed here.
In the figure, the water flowing in the pipe (flow in) comes with section 1 and flow out goes with section 3.
The several rotations are:
- A1 = Inlet area in m2.
- D1 = Diameter of Inlet.
- D2 = Diameter of the throat.
- A2 = Throat area in m2.
- P1 = Pressure at the inlet in N/m2.
- P2 = Pressure at the throat in N/m2.
- V1 = Velocity at inlet in m/sec
- V2 = Velocity at throat in m/sec.
- h = Pressure heads.
- Cd = Coefficient of Discharge. This is unitless.
- Qact= Actual discharge in m3/sec.
- Qthe= Theoretical discharge in m3/sec.
The other way to find h (Pressure heads) by using differential U–Tube Manometer:
The liquid in the manometer is heavier than the flowing fluid in the pipe.
- Sh =Specific gravity of the heavier liquid.
- x = Difference of the heavier liquid column in U-tube
- S0 =The Specific gravity of flowing fluid.
- Sl =Specific gravity of the lighter liquid.
h = x [ (Sh / S0) – 1]
The liquid in the manometer is lighter than the flowing fluid in the pipe.
h = x [1- (Sl / S0) ]
Preparing for Competitive Exam? If Yes, then must go for this book:
The most valuable and trusted Mechanical Objective book by R S Khurmi 5000+ Objective.
Venturi Meter Application / Uses:
The Venturi Meter application are:
- Calculating the flow rate of fluid that is Discharge in the pipe.
- The flow of chemical in pipelines.
- This is widely used in large-diameter pipes such as found in the waste treatment process.
- Also used in the medicine department for the measure, the flow rate of blood in arteries.
- This also used where high-pressure recovery is required.
Venturi Meter Advantages / Merits:
The Venturi Meter advantages are:
- Power loss is very less.
- This can be used where a small head is available.
- High reproducibility (the extent to which consistent results are obtained when an experiment is repeated).
- Accuracy is high over wide flow ranges.
- This can also be used for a compressible and incompressible fluid.
- This device is easy to operate.
- The coefficient of discharge (Cd ) for the venturi meter is high.
- This is widely used for high flow rate (Discharge).
Venturi Meter Disadvantages / Demerits / Limitation:
The Venturi Meter disadvantages are:
- The installation cost of a venturi meter is high.
- There are little difficulties while maintenance.
- This device can not be used where the pipe has a small diameter of 76.2 mm.
- This system occupies more space as compared to the orifice meter.
- It has a limitation of lower Reynolds number of 150,000.
- It is expensive and a little bulky.
Venturi Meter PDF:
Download PDF from Facebook or Telegram group.
Join the Facebook group: Engramritkumar Facebook
Connect Telegram group: Engramritkumar Telegram
Join WhatsApp group: Engramritkumar WhatsApp
Today we have learned about Definition, Parts, Working Principle, Diagram, Experiment, Derivation, Formula, Application, Advantages, Disadvantages, PDF. If I have missed something, kindly inform me through commenting. If you like this article, don’t forget to share it on social networks. Subscribe our website for more informative articles. Thanks for reading it. Thank you.
Other Top Trending Post in details:
- Production Machine:
- Ignition System:
- Vehicle System:
- Welding and It’s Type
Help others to read by share this Post